For more information about this product please call +44 (0) 2392 488 240 or email us using the web form below:
Salinometer calibration for measurement of oceanic samples is traditionally carried out using IAPSO Standard Seawater (S = 35). For measurement of samples of higher and lower salinity, OSIL produce standards at salinities of 10, 30 and 38 which are all available in the Linearity Pack.
Salinometer linearity checks provide a useful diagnostic tool in assessing the function of a salinometer and are routinely carried out at the OSIL Service Centre. Offset values can be used to improve the accuracy of measurements made away from 35 salinity. A large offset away from the 35 salinity point can be indicative of electronics misalignment, cell ageing or temperature bath malfunction.
A typical procedure for linearity correction would be:
1) Standardise the salinometer with P-series Normal Standard Seawater (S = 35).
2) Measure the salinities for 38H-series, 30L-series and 10L-series Standard Seawaters.
3) Calculate the difference between the measured value and the label value for each of these standards i.e. measured value-label value = salinometer offset.
4) Plot a graph of offset at each salinity (a linearity curve).
5) Use the plotted offset data to correct salinity measurements away from salinity 35 e.g. at salinity 10 the offset is -0.002 in salinity, therefore add 0.002 to your measured values around 10.
Guildline Autosal accuracy specified as better than +/- 0.002 in salinity over the working range (2-42)
Guildline Portasal accuracy specified as better than +/- 0.003 in salinity over the working range (2-42)
View our Seawater Standards Brochure for further details on all the standards available from OSIL.
Low Nutrient Seawater